The QuantiVirus™ SARS-CoV-2 Test Kit is based on Real-Time PCR technology, developed for specific detection of SARS-CoV-2 (COVID-19) viral RNA extracted from nasopharyngeal swabs, oropharyngeal swabs and sputum. The sensitivity is 100 copies per mL of SARS-CoV-2 viral with a 95% confidence. The assay is highly specific without cross-reaction with different types of non-SARS-CoV-2 species tested. The clinical validation shows 96.7% sensitivity and 100% specificity.
For Emergency Use Authorization (EUA) only. For prescription use only. For in vitro diagnostic use.
Looking for a higher throughput kit? Learn more about our newly launched QuantiVirus™ SARS-CoV-2 Multiplex Test Kit.
Key Product Information
- FDA Emergency Use Authorization (EUA) Approved
- Detects three genes, Orf1ab, N and E genes
- Detects different SARS-CoV-2 virus strains
- Sensitivity is 100 copies per mL
- Less than 2 hours from RNA to results on most qPCR machines
Nasopharyngeal Swabs, Oropharyngeal Swabs and Sputum
Thermo Fisher (ABI) QuantStudio 5, Thermo Fisher (ABI) 7500 Fast Dx, and Bio-Rad CFX 384
GET A QUOTE NOW
Performance Characteristics of the QuantiVirus™ SARS-CoV-2 Test Kit
100 copies per mL of SARS-CoV-2 viral with a 95% confidence
Low Sample Volume
only 2 uL viral RNA sample
Detection of three target genes for calling positive results
Wide Dynamic Range
Allowing detection of from 10 million copies to single copy analyte in linear range
Coronavirus (SARS-CoV-2) Detection Tests are Crucial
The global outbreak of novel coronavirus, SARS-CoV-2 (COVID-19), caused over 3000 people’s death and huge economic damage. The number is still increasing. Detecting the virus early is essential to contain the virus from infecting more people. However, in the United States, the unavailability of enough reliable detection kits in time has caused concerns and panic in the society.
*The numbers are calculated on March 9, 2020 by https://www.worldometers.info/coronavirus/
Without the test, the general public does not know if they get the novel coronavirus or the regular flu virus as they share similar symptoms
Without the test, the general public does not know if they carry the virus without symptoms and further infect their closed ones and others who have contact with them
Without the test, the general public does not know if they carry the virus and miss the treatment window
To meet the unmet medical needs, DiaCarta has developed the QuantiVirus™ SARS-CoV-2 Test Kit to accurately detect the virus with high sensitivity using nasopharyngeal swabs, oropharyngeal swabs and sputum.
How Does the QuantiVirus™ SARS-CoV-2 Test Kit Work?
Figure 1. Amplicon Target on SARS-CoC-2 genome. E: envelope protein gene; M: membrane protein gene; N: nucleocapsid protein gene; ORF: open reading frame; RdRp: RNA-dependent RNA polymerase gene; S: spike protein gene.
The QuantiVirus™ SARS-CoV-2 Test Kit is a qRT-PCR assay for the qualitative detection of SARS-CoV-2 viral RNA.
Three genes of the SARS-CoV-2 including N, Orf1ab and E are targeted in the qRT-PCR assay and primers and Taqman probes are designed in the conserved region of the SARS- CoV-2 virus-specific genome region to allow sensitive and specific amplification and detection of the virus. The human RNase P gene is used as internal and extraction control to monitor viral RNA extraction efficacy and assess amplifiable RNA/DNA in the samples to be tested.
The assay has a wide dynamic range as shown in the figure below: Amplification curve of 10 -fold serial dilution of templates showing the dynamic range.
The assay performance summary for QuantiVirus™ SARS- CoV-2 Test Kit
- Intra-assay reproducibility: Coefficient of Variation (%): < 3%
- Operator reproducibility: Coefficient of Variation (%): < 1.5%
- Inter-instrument reproducibility: Coefficient of Variation (%): < 5%
- Assay detection limit (LOD): 1 copy of virus RNA or 100 copies per ml sample
Clinical Evaluation (In Vitro Transcribed Viral RNA Spiking into Sputum)
Clinical evaluation of the QuantiVirus™ SARS-CoV-2 Test kit was conducted with contrived sputum specimens including 30 positive and 30 negative samples. 20 sputum samples were contrived with RNA (in vitro transcripts) templates at 2xLOD (2×100 copies/mL) and 10 sputum samples were contrived with different higher RNA (500, 1000, 5000 and 10000 copies/uL). 30 specimen were contrived only with water as the negative control. The contrived clinical samples were tested blindly to generate the Positive Predictive Value (PPV), Negative Predictive Value (NPV) as a measurement of estimated Diagnostic Accuracy.
Data show that there is a 95% agreement with the spiking sample with 2X LOD (2×100 copies/mL), but one sample cannot be detectable. With higher concentration spiking samples (500, 1000, 5000 and 10000 copy/uL), 10 out of 10 samples were detectable. For negative control, Thirty samples were negative.
Table: Contrived clinical sample evaluation with in vitro transcribed RNA
The above table shows:
Clinical Sensitivity = 96.7% (95% CI, 81.4%-99.8%)
Clinical Specificity = 100% (95% CI, 85.4%-100%)
Positive Predictive Value (PPV) = 100% (95% CI, 85.8%-100%
Negative Predictive Value (NPV) = 96.6% (95% CI, 80.9%-99.8%)
Third-Party Clinical Test Results
The third-party clinical test results demonstrated that DiaCarta’s QuantiVirus™ SARS-CoV-2 Test Kit shared the same testing results (100% match), compared with peer products, even at much lower sample concentration in some cases.
|Sample ID||Abbott m2000||CDC Assay||DiaCarta's QuantiVirus™ Assay|
|A||Detected||Detected at 1:100 dilutions|
|B||Detected||Detected at 1:1000 dilutions|
|C||Not Detected||Not Detected||Not Detected|
|D||Not Detected||Not Detected||Not Detected|
|E||Not Detected||Not Detected||Not Detected|
|F||Not Detected||Not Detected||Not Detected|
|G||Not Detected||Not Detected||Not Detected|
|H||Not Detected||Not Detected||Not Detected|
|I||Not Detected||Not Detected||Not Detected|
|N||Not Detected||Not Detected||Not Detected|
Streamlined Workflow for the QuantiVirus™ SARS-CoV-2 Test Kit
Extract RNA from patient sample and extraction control using Thermo Fisher PureLink viral RNA/DNA mini kit or Qiagen QIAamp® Viral RNA Mini Kit
Set up the assay reactions using primer/probe mixes, and one-step RT-qPCR master mix. A positive (PC), NTC (NC)and extraction control (EC) must be included for every run
Perform RT-qPCR using the Applied Biosystems™ QuantStudio 5, or 7500 Dx or BioRad CXF 384 Real-Time PCR Instrument
Data analysis and review results interpretation for patient samples
Comparison of QuantiVirus™ SARS-CoV-2 Test Kit and QuantiVirus™ SARS-CoV-2 Multiplex Test Kit
|QuantiVirus™ SARS-CoV-2 Test (Presented on the current web page)||QuantiVirus™ SARS-CoV-2 Multiplex Test|
|Sample types||Sputum, NP, OP||Sputum, NP, OP, and saliva|
|Ease of use||One sample in multiple qPCR reaction wells (4 assay mixes/per sample)||One sample in one qPCR reaction well (1 assay mix/per sample|
|Throughput||21 samples (96-well plate) or 93 samples (384-well plate) per run||93 samples (96-well plate) or 381 samples (384-well plate) per run|
|Sensitivity||Detects as low as 100 copies of Viral RNA /mL sample||Detects as low as 50 copies of Viral RNA /mL sample|
|Specificity||Proven by in silico analysis as well as wet lab testing||Proven by in silico analysis as well as wet lab testing|
|Sample input||(2 - 5.5 µL)x4/per sample||2 - 5.5 µL per sample|
|Validated instrument||Validated on ABI 7500 Fast Dx, ABI QuantStudio 5 and Bio-Rad CFX 384||Validated on ABI 7500 Fast Dx, ABI QuantStudio 5 and Bio-Rad CFX 384|
|Turnaround time||Less than 2 hours from RNA extraction to results||Less than 2 hours from RNA extraction to results|
|Detected genes||Orf1ab, N and E genes||Orf1ab|
|Registration status||CE/IVD Marked|
FDA EUA Approved
FDA EUA Filed
Product Specifications and Order Information
US Customer Ordering Information (FDA EUA Approved)
Product Name: QuantiVirus™ SARS-CoV-2 Test Kit
24-Reaction Kit Catalog Number: DC-11-0007
48-Reaction Kit Catalog Number: DC-11-0008
480-Reaction Kit Catalog Number: DC-11-0009
Outside US Customer Ordering Information (CE-Marked)
Product Name: QuantiVirus™ Real-Time PCR Coronavirus (SARS-CoV-2) Detection Test
Outside the US, this product is available for diagnostic use (CE-marked)
24-Reaction Kit Catalog Number: DC-11-0005E
48-Reaction Kit Catalog Number: DC-11-0004E
Nasopharyngeal Swabs, Oropharyngeal Swabs and Sputum
Thermo Fisher (ABI) QuantStudio 5, Thermo Fisher (ABI) 7500 Fast Dx, Bio-Rad CFX 384
Stable for 12 months at -25°C to -15°C